Portable steam sterilizer is a sterilization boiler with shutter-type pressure vessel safety interlock device, independent electrical instrument control system, automatic constant temperature, heat preservation and timing. It is made of stainless steel material. Beautiful, hygienic, safe and durable.
Check the equipment: Check whether the portable steam sterilizer power cord, steam exhaust pipe, and drainage pipe are in good condition, and whether the water discharge valve and steam discharge valve are closed.
Portable steam sterilizer is a common sterilization and sterilization equipment in the laboratory. It has important applications in many industries. It has the advantages of large heating area, high thermal efficiency, uniform heating, short boiling time of liquid material, and easy control of heating temperature.
The actual working temperature and pressure of the portable steam sterilizer are important factors that affect the sterilization effect. As a high-risk device for clinical application, it is very important to carry out regular quality inspection and calibration.
Autoclaving refers to the use of thermal energy to kill microorganisms. The principle is that high temperature denatures proteins in the protoplasm, inactivates enzymes, and causes the death of microorganisms.
In the airtight high-pressure environment of the portable steam sterilizer, the steam in it cannot escape, and the pressure continues to rise, so that the boiling point of the water continues to increase, so the temperature in the pot also increases.
Insert the portable steam sterilizer in china power plug into a 10A/220V external power socket. Turn the power switch to the "on" position, the power indicator light is on, and the electric heating tube starts to heat.